I figured it might be a good idea to bring this one up on the blog, just in case. Over the weekend, I finally got my flow test rig setup for my thesis. The rig is pretty similar to the fuel half of our igniter cart–basically it’s just a fluid tank with a regulated pressure input, some fill plumbing, and a valve between it and the water jet nozzle. I should have the linear amplifier for driving the piezoelectric crystal fairly soon (Charles Pooley of Microlaunchers is helping me out with the design–he’s very good at analog electronics design).
The experiment I need to run involves using high-speed photography techniques to determine if the pulsed nozzle is creating a pulsed flow. In order to have a clear comparison, I’m also doing some runs with no driving voltage on the piezoelectric crystal, basically as a control experiment. The smoother and more stable and steady I can get the unpulsed jet, the more obvious even the slightest fluctuations are going to be for the pulsed case.
I was able to enlist the help of XCOR’s Mike Massee for help with the photography portion, but we quickly realized that the nice strobes he had were far too slow for the task at hand. Those strobes were mostly for the entertainment business, so they had a pulse duration of about 250 microseconds. The problem is that I’m trying to detect oscillations on an 0.33mm jet that’s traveling at nearly 60-80m/s when it leaves the nozzle, and in 250 microseconds, the jet travels nearly 15mm (or about 45 jet diameters)! I ran the numbers, and in order to prevent the picture of the jet from completely blurring, I need to have the jet move less than 33 microns (about a tenth of the jet diameter) during the flash. The problem is that this implies a pulse duration of around 0.5 microseconds.
That’s really darned fast.
I discovered while doing some googling after our test run that the strobe I had at BYU that I thought was so crappy actually had a minimum pulse duration in the 0.5-0.8 microseconds range, and that that was actually on the high end for stroboscopes.
Here’s my problem. The thesis draft has to be far enough along that I can get my advisor to agree that I’m ready to schedule my defense by the 18th. This means that the thing is basically written and that I’m in the last minute polishing phase. The experimental validation is very important, and to drop that at the last second will make it a lot harder for me to get my thesis accepted. Since it turns out that the strobes we have just aren’t even close to what I need to pull this project off, I need to come up with another option. That means I need to come up with either a high speed camera setup, or a high speed strobe (with pulse durations less than 1 microsecond) within the next week.
The company that makes the strobes that I used at BYU (They’re General Radio Strobotac strobes) wants almost $6k for the strobes, and doesn’t appear to rent them. There are a couple of places that sell “nano-pulse” systems that use xenon spark flashbulbs to create a flash of light that lasts less than 100 nanoseconds. Those would be even better than a Strobotac (due to the shorter pulse duration), but as far as I couldn’t find any of those for rent, or for sale used on ebay. It is possible to rent high speed cameras, but they tend to run several hundred to a few thousand dollars just for a day or two, which is out of my price range.
So, my question is, does anyone who reads this know a way I could get my hands on something like that within the next week? Anyone have any ideas?
Latest posts by Jonathan Goff (see all)
- Administrivia - July 17, 2018
- Research Papers I Wish I Could Con Someone Into Writing Part I: Lunar ISRU in the Age of RLVs - March 9, 2018
- Random Thoughts: A Now Rather Cold Take on BFR - February 5, 2018